The SediMeter instrument measures a vertical turbidity profile through the bottom of lakes, rivers, or the sea, in order to monitor sedimentary processes. Since it uses light (near infra-red) it has to stay reasonably clean to function properly, something that can be a challenge in some environments. For that reason we have version SM3B with a built-in mechanical cleaner. Apart from being costly, this has the disadvantage of creating a possibility for failure, by having a moving part in a liquid that sometimes is full of suspended sand particles. A method of keeping clean without moving parts is needed.
The SM3C model that we now announce has a vibrator â€” as in a mobile phone â€” that shakes briefly but intensely at preset intervals. The video shows the effect: particles and bubbles dislodged from the sensor.
Another novelty of 2017 is that we are doubling the resolution by measuring both straight backscatter and oblique backscatter. We will return to that topicÂ in a later post since it has wide-ranging implications for the usefulness of the instrument.
Much more to come in 2017, never before have we have so many new features in the pipeline!